<?xml version="1.0" encoding="UTF-8"?>
<!DOCTYPE ep-patent-document SYSTEM "OpenDataPatent_cz.dtd">
<ep-patent-document lang="en" file="OpendataIPOCZ_PT_DIFF_PUV2014-29498.xml" country="CZ" doc-number="27510" kind="U1">
  <bibliographic-data country="CZ">
    <publication-reference>
      <document-id>
        <country>CZ</country>
        <doc-number>27510</doc-number>
        <kind>U1</kind>
        <date>2014-11-20</date>
      </document-id>
    </publication-reference>
    <classifications-ipcr>
      <classification-ipcr sequence="1">
        <section>G</section>
        <class>01</class>
        <subclass>J</subclass>
        <main-group>3</main-group>
        <subgroup>00</subgroup>
        <ipc-version-indicator>
          <date>20060101</date>
        </ipc-version-indicator>
        <classification-level>A</classification-level>
        <symbol-position>F</symbol-position>
        <classification-value>I</classification-value>
        <action-date>
          <date>20140710</date>
        </action-date>
        <classification-status>B</classification-status>
        <classification-data-source>H</classification-data-source>
        <generating-office>
          <country>CZ</country>
        </generating-office>
      </classification-ipcr>
      <classification-ipcr sequence="2">
        <section>H</section>
        <class>03</class>
        <subclass>M</subclass>
        <main-group>1</main-group>
        <subgroup>00</subgroup>
        <ipc-version-indicator>
          <date>20060101</date>
        </ipc-version-indicator>
        <classification-level>A</classification-level>
        <symbol-position>L</symbol-position>
        <classification-value>I</classification-value>
        <action-date>
          <date>20140710</date>
        </action-date>
        <classification-status>B</classification-status>
        <classification-data-source>H</classification-data-source>
        <generating-office>
          <country>CZ</country>
        </generating-office>
      </classification-ipcr>
      <classification-ipcr sequence="3">
        <section>H</section>
        <class>03</class>
        <subclass>M</subclass>
        <main-group>1</main-group>
        <subgroup>08</subgroup>
        <ipc-version-indicator>
          <date>20060101</date>
        </ipc-version-indicator>
        <classification-level>A</classification-level>
        <symbol-position>L</symbol-position>
        <classification-value>I</classification-value>
        <action-date>
          <date>20140710</date>
        </action-date>
        <classification-status>B</classification-status>
        <classification-data-source>H</classification-data-source>
        <generating-office>
          <country>CZ</country>
        </generating-office>
      </classification-ipcr>
      <classification-ipcr sequence="4">
        <section>H</section>
        <class>03</class>
        <subclass>M</subclass>
        <main-group>1</main-group>
        <subgroup>06</subgroup>
        <ipc-version-indicator>
          <date>20060101</date>
        </ipc-version-indicator>
        <classification-level>A</classification-level>
        <symbol-position>L</symbol-position>
        <classification-value>I</classification-value>
        <action-date>
          <date>20140710</date>
        </action-date>
        <classification-status>B</classification-status>
        <classification-data-source>H</classification-data-source>
        <generating-office>
          <country>CZ</country>
        </generating-office>
      </classification-ipcr>
      <classification-ipcr sequence="5">
        <section>H</section>
        <class>03</class>
        <subclass>F</subclass>
        <main-group>1</main-group>
        <subgroup>26</subgroup>
        <ipc-version-indicator>
          <date>20060101</date>
        </ipc-version-indicator>
        <classification-level>A</classification-level>
        <symbol-position>L</symbol-position>
        <classification-value>I</classification-value>
        <action-date>
          <date>20140710</date>
        </action-date>
        <classification-status>B</classification-status>
        <classification-data-source>H</classification-data-source>
        <generating-office>
          <country>CZ</country>
        </generating-office>
      </classification-ipcr>
      <classification-ipcr sequence="6">
        <section>H</section>
        <class>03</class>
        <subclass>F</subclass>
        <main-group>1</main-group>
        <subgroup>00</subgroup>
        <ipc-version-indicator>
          <date>20060101</date>
        </ipc-version-indicator>
        <classification-level>A</classification-level>
        <symbol-position>L</symbol-position>
        <classification-value>I</classification-value>
        <action-date>
          <date>20140710</date>
        </action-date>
        <classification-status>B</classification-status>
        <classification-data-source>H</classification-data-source>
        <generating-office>
          <country>CZ</country>
        </generating-office>
      </classification-ipcr>
    </classifications-ipcr>
    <application-reference>
      <document-id>
        <country>CZ</country>
        <doc-number>2014-29498</doc-number>
        <date>2014-04-23</date>
      </document-id>
    </application-reference>
    <language-of-filing>cs</language-of-filing>
    <language-of-publication>cs</language-of-publication>
    <priority-claims>
      <priority-claim sequence="1" kind="national">
        <country>CZ</country>
        <date>2014-04-23</date>
      </priority-claim>
    </priority-claims>
    <invention-title lang="CS">Uspořádání dvouparametrického spektrometrického systému</invention-title>
    <invention-title lang="EN">Arrangement of two-parameter spectrometric system  </invention-title>
  </bibliographic-data>
  <description>
    <heading>

Uspořádání ďvouparametrického spektrometrického systému</heading>
    <heading>

Oblast techniky</heading>
    <p id="p0001" num="0001">Technické řešení se týká uspořádání analogově-digitálního systému pracujícího s frekvencí inter-ních synchronizačních signálů v hodnotách desítek gigahertzů (GHz). Digitální systém je určen pro dvouparametrická nebo jednoparametrická spektrometrická měření.</p>
    <heading>

Dosavadní stav techniky</heading>
    <p id="p0002" num="0002">Běžně vyráběné jednoparametrické impulsní spektrometrické systémy záření slouží k detekci energie dopadajících částic. Skládají se z analogové a číslicové části. Analogová část přizpůso-buje výstup detektoru záření další části měřícího řetězce, tvaruje vhodně impuls nesoucí informaci o energii, kterou detekovaná částice deponovala v detektoru atd. V další části - v analog/číslo převodníku (dále AD převodníku) - se amplituda analogového signálu konvertuje na číslo, které je vstupem pro následné určení energie detekované částice. Délka náběžné nebo se-stupné hrany výstupního impulzu, nesoucí informaci o typu detekované částice, není jednopara-metrickými impulsními spektrometrickými systémy záření vyhodnocována.</p>
    <p id="p0003" num="0003">Doposud vyráběné a dostupné dvouparametrické spektrometrické systémy vyhodnocují druhý parametr, nesoucí informaci např. o typu detekované částice, poměrně složitými a ne příliš rych-lými analogovými obvody. Tyto analogové obvody využívají rozdílného časového průběhu odezvy scintilátoru na dopad různých typů částic k jejich rozlišení (Pulse Shape Discrimination). Tvar impulzu vystupujícího z vhodného scintilátoru zesílený fotonásobičem je tedy odlišný pro jednotlivé druhy částic jaderného záření - rozdíly se však pohybují v jednotkách nanosekund. Z tohoto důvodu analogové aparatury používají metod založených na integraci vzestupné a se-stupné části odezvy na výstupu scintilátoru-fotonásobiče a následném porovnání takto získaných veličin. Metod diskriminace jek dispozici více, ale všechny mají nedostatky - nemohou vyhod-nocovat překrývající se výstupní impulzy, jsou rozměrné, mají značný příkon, použití mimo labo-ratorní prostředí je prakticky nemožné a především umožňují zpracovat pouze nízké četnosti impulsů pohybující se kolem hodnoty 5 tisíc impulzů za sekundu. Digitální zpracování informací <sub>0</sub> signálech s výše uvedenými parametry je limitováno dynamickými parametry použitých elek-tronických součástek a použitím vhodných algoritmů pro separaci tvarem velmi podobných im-pulzů. Nejmodernější číslicové obvody umožňují konstruovat sestavy schopné pracovat se vzor-kovacími kmitočty v řádech stovek MHz (megahertzů).</p>
    <heading>

Podstata technického řešení</heading>
    <p id="p0004" num="0004">Cílem technického řešení je předložit uspořádání analogově-digitálního systému, který by umož-ňoval digitálně zpracovávat oba parametry měřených signálů a tak podstatně zrychlil a zpřesnil měření.</p>
    <heading>

Pojmy uváděné v následujícím textu mají tento význam:</heading>
    <p id="p0005" num="0005">1 - First In, First Out, metoda organizace záznamu do paměti „první dovnitř, první ven; 2 - transmitter - název výstupního bodu vysílače dat; 3 - receiver - název vstupního bodu přijímače dat;</p>
    <p id="p0006" num="0006">4- standard systémové sběrnice číslicových systémů (náhrada starších standardů PCI, PCI-X a</p>
    <p id="p0007" num="0007">AGP);</p>
    <p id="p0008" num="0008">5- standard pro rozšíření interfejsu XGMII (10 Gigabit Media Independent Interface) mezi vrstvami MAC a PHY standardu komunikačního protokolu 10 Gigabit Ethernet (lOGbE).</p>
    <p id="p0009" num="0009">Výše uvedené nedostatky jsou odstraněny upořádáním analogově-digitálního systému, jehož podstata spočívá v tom, že má blok detektoru a zesilovačů, zahrnující detektor a předzesilovač, připojen na blok převodníků, zahrnující první a druhou dvojici AD převodníků, které jsou pomocí datových sběrnic připojeny blok úpravy dat zahrnující časovač, který je dále přes datové sběrnice připojen na blok interface, zahrnující převaděč, který je napojen na server zahrnující <!-- EPO <DP n="2"> -->
operační paměť typ RAM (RAM - random-access memory, paměť s libovolným výběrem) a pevný disk.</p>
    <p id="p0010" num="0010">Podstatou technického řešení je, že výstupní signál fotonásobiče je AD převodníkem bezpro-středně převáděn do digitální podoby a všechny ostatní operace sloužící pro rozlišení typu impulzu a určení energie registrované částice jsou prováděny číslicově. Číslicové obvody analo-gově-digitálního systému provádí:</p>
    <heading>

a) filtrace šumu,</heading>
    <heading>

b) spouštění AD převodníku na počáteční hranu signálu,</heading>
    <heading>

c) komprimaci naměřených dat,</heading>
    <heading>

d) přenos dat do vnější paměti (pevného disku) serveru.</heading>
    <p id="p0011" num="0011">Navrhovaný postup umožní sestavit energetické spektrum měřeného záření, což není funkce po-skytovaná standardními spektrometrickými systémy a v rozsahu, který žádné, v současné době dostupné zařízení, není schopno poskytnout.</p>
    <p id="p0012" num="0012">Předností navrženého zařízení je rychlost zpracování dat (řádově o dva řády rychlejší záznam dat ve srovnání s dostupnými zařízeními s podstatně horšími užitnými vlastnostmi), velký dynamický rozsah zpracovávaných vstupních signálů, nízká spotřeba a hmotnost, možnost použití v provoz-ních podmínkách a v terénu.</p>
    <p id="p0013" num="0013">Výhodou též je nízká spotřeba, která umožňuje zařízení napájet měničem z baterií nebo akumu-látorů mimo distribuční síť 230 V.</p>
    <p id="p0014" num="0014">Použité elektronické komponenty umožnily redukci technického vybavení spektrometrického řetězce na minimum, což je dáno možností převodu analogového signálu na výstupu detektoru extrémně rychlým analogově číslicovým převodníkem a vysokou rozlišovací schopností.</p>
    <p id="p0015" num="0015">Další variantou uspořádání analogově-digitálního systému je duální zapojení, bloku úpravy dat, kdy každý z AD převodníků (v použitém obvodu ADC12D1000 jsou dva převodníky) je spouš-těn samostatnými synchronizačními signály navzájem posunutými o 0,5 nanosekundy, čímž je dosaženo dvojnásobné vzorkovací frekvence. Ve výše jmenovaném duálním zapojení analogově-digitálního systému blok úpravy dat zahrnuje časovač připojený k bloku zpracování dat, který je připojen k bloku FIFO<sup>1</sup> filer (plnič - obvod zajišťující funkci FIFO), jenž je připojen k bloku generování paketů, jenž je přes výstup TX<sup>2</sup> připojen na vstup RX<sup>3</sup> bloku přepínače, jenž je přes výstup TX připojen na vstup RX bloku interface, jehož výstup TX je propojen na vstup RX bloku řídicí jednotky/řadiče, který má výstup TX připojen na vstup RX bloku přepínače, zatímco blok generování paketů je připojen přes blok přepínače k bloku řídicí jednotka/řadič pro řízení přenosu dat, přičemž blok řídicí jednotka/řadič je dále připojen k blokům a AD převodníky pro jejich řízení a k bloku interface.</p>
    <p id="p0016" num="0016">Výhodou je, že veškeré další zpracování a přenosy dat se provádí v digitálním tvaru. Součástí popisovaného dvouparametrického spektrometrického systému (předmětu průmyslového vlastnictví - PPV) je obvod FPGA (Field Programmable Gate Array, programovatelné hradlové pole), který je konfigurován pro filtrování neužitečných úseků měřeného signálu a pro předzpracování naměřených dat.</p>
    <p id="p0017" num="0017">Realizované řešení umožňuje miniaturizaci spektrometrických aparatur, zvýšení odolnosti proti rušení, podstatné zvýšení četnosti zpracovávaných odezev senzoru jaderného záření (zkracování a zlevňování měření) a taktéž filtraci či zpracování superponovaných odezev registrovaných čás-tic jaderného záření. V dvouparametrickém režimu sloužit jako spektrometr neutronů a záření gama s organickým scintilátorem stilben, proporcionálním detektorem plněným vodíkem nebo kapalným scintilátorem pro separaci a následným zpracováním nebo záznamem odezev fotonů a neutronů. V jednoparametrickém režimu může být k měření spekter použit polovodičový detektor typu HPGe (high purity germanium - monokrystal vysoce čistého germánia) nebo scintilační detektory typu LaBr (lanthanbromit - chemický vzorec LaBr<sub>3</sub>), Nal(Tl) (jodid sodný aktivovaný thaliem), BGO (Bismuth germanium oxide - chemický vzorec BÍ<sub>4</sub>Ge<sub>3</sub>0i<sub>2</sub>) apod.</p>
    <!-- EPO <DP n="3"> -->
    <heading>

Přehled obrázků na výkresech</heading>
    <p id="p0018" num="0018">Příkladné provedení tohoto technického řešení je znázorněno na přiložených výkresech, kde na obr. 1 je znázorněno blokové schéma analog-digitálního systému a na obr. 2 je znázorněno blokové schéma bloku úpravy dat (FPGA).</p>
    <heading>

Popis příkladného provedení</heading>
    <p id="p0019" num="0019">Upořádání analogově-digitálního systému podle tohoto technického řešení bude osvětleno v následujícím popisu na příkladném provedení s odkazem na příslušné výkresy.</p>
    <p id="p0020" num="0020">Základní uspořádání zapojení analogově-digitálního systému je znázorněno na obr. 1. Blokové schéma analog-digitálního systému obsahuje blok l detektoru a zesilovače, který zahrnuje, známým způsobem propojené následující součástky, spektrometrický senzor JJ_ (např. stilbene), jenž reaguje na dopadající směsné záření záblesky ve viditelném světelném spektru, fotonásobič mě-nící světelný záblesk na napěťový impulz a předzesilovač 12, který přizpůsobuje impedance pro-pojovacího kabelu mezi výstupem fotonásobiče a vstupními obvody AD převodníků. Dále je blok I detektoru a zesilovače připojen na blok 2 AD převodníků, jenž zahrnuje první dvojici AD pře-vodníků 21 a druhou dvojici AD převodníků 22. Blok 2 AD převodníků zajišťuje převod analogového signálu z bloku 1 detektoru a zesilovačů, do digitálního tvaru. Na desku je možno umístit dva integrované obvody ADC1 a ADC2, každý s dvojicí rychlých AD převodníků 21 a 22, je-jichž řízení zajišťuje blok 3 úprav dat (realizovaný například obvodem FPGA), a které jsou pomocí datových sběrnic připojeny k bloku 3 úpravy dat, zahrnující časovač 3J_. V bloku 3 úpravy dat jsou instalovány algoritmy pro řízení AD převodníků, filtraci šumu na výstupu AD převod-níků, algoritmy předzpracování digitálních vzorků vstupního analogového signálu, dále algoritmy řízení interní vyrovnávací paměti, například typu FIFO, která umožňuje přizpůsobit rychlost generování vzorků vstupního signálu rychlosti datového kanálu připojeného serveru, algoritmy předávání dat do bloku 4 interface. V použitém obvodu FPGA jsou dále implementovány následující obvody a to: - vstupní obvody datových linek od AD převodníků pracující s rychlostí 24 Gb/s (gigabitů za sekundu) pro každý z AD převodníků,</p>
    <heading>

- vyrovnávací paměť dat typu FIFO,</heading>
    <p id="p0021" num="0021">- výstupní obvody datových linek pro přenos dat do datového rozhraní serveru dvouparametrického spektrometrického systému pracující s rychlostí 10 Gb/s.</p>
    <p id="p0022" num="0022">Blok 3 úpravy dat je dále přes datové sběrnice připojen na blok 4 interface, zahrnující převaděč 41, jedná se o datové rozhraní realizované deskou PCI-EXPRESS<sup>4</sup> umožňující rychlost přenosu 12 Gb/s, do bloku 5 server.</p>
    <p id="p0023" num="0023">Blok 4 interface/Combo card je napojen známým způsobem na server 5 zahrnující operační pa-měť RAM 5J_ serveru a pevný disk 52 (externí paměť) serveru. Jeho kapacita je zvolena tak, aby umožnila uložení takového množství, které poskytne statisticky věrohodné podklady pro vyčís-lení energetického spektra měřeného směsného záření. Na disku se ukládají komprimovaná data z bloku 3 úpravy dat, s obvodem FPGA. Tato data jsou dále zpracovávána separačními algoritmy, které zajistí jednak identifikaci typu detekované částice (prioritně neutronu nebo gama částice) a jednak její energii.</p>
    <p id="p0024" num="0024">Varianta uspořádání analogově-digitálního systému s duálním zapojením, bloku 3 úpravy dat/FPGA je znázorněna na obr. 2. Její zapojení je následující. Blok 3 úpravy dat zahrnuje časo-vač 3J_ připojený k bloku 32 zpracování dat, který je připojen k bloku 33 FIFO filer, jenž je při-pojen k bloku 34 generátoru paketů, jenž je přes výstup TX připojen na vstup RX bloku 35 přepí-nače, jenž je přes výstup TX připojen na vstup RX bloku 36 interface, jehož výstup TX je propojen na vstup RX bloku 37 řídicí jednotky/řadiče, který má výstup TX připojen na vstup RX bloku 35 přepínače, zatímco blok 34 generování paketů je připojen přes blok 35 přepínače k bloku 37 řídicí jednotka/řadič pro řízení přenosu dat, přičemž blok 37 řídicí jednotka/řadič je dále připojen k blokům 2J_ a 22 AD převodníky pro jejich řízení a k bloku 36 interface.</p>
    <!-- EPO <DP n="4"> -->
    <p id="p0025" num="0025">V této části jsou uvedeny algoritmické a funkční moduly bloku 2 AD převodníků bloku 3 úpravy dat s použitým obvodem FPGA a bloku 4 interface/Combo card.</p>
    <heading>

Blok 2 analogově digitálních převodníků ADC, zahrnuje:</heading>
    <p id="p0026" num="0026">- řídicí registry AD převodníků pro nastavení režimu práce první a druhé dvojice AD převod-níků 2J_ a 22.</p>
    <heading>

Blok 3 úpravy dat s použitým obvodem FPGA zahrnuje:</heading>
    <p id="p0027" num="0027">Blok 3J_ ADC data and clock synchronization (časování vzorkování jednotlivých AD převod-níků),</p>
    <p id="p0028" num="0028">Blok 32 Data processing area (oblast pro implementaci komprimačních a filtračních algoritmů),</p>
    <heading>

Blok 33 FIFO filler asynchronous FIFO. Zajišťuje funkce:</heading>
    <heading>

- plnění fronty dat,</heading>
    <p id="p0029" num="0029">- generování informací o jednotlivých paketech ukládaných do FIFO pamětí (časové razítko, číslo impulzu, data, zabezpečení dat),</p>
    <heading>

Blok 34 Generátor paketů pro přenos dat po sběrnici PCI-EXPRESS,</heading>
    <p id="p0030" num="0030">Přepínač 35 (například XAUI<sup>5</sup> switcher podle standardu 10 Gigabit Media Independent Interface),</p>
    <p id="p0031" num="0031">Interface 36 XAUI pro komunikaci s blokem 4 interface/Combo Card (přizpůsobení toku dat pro přenos dat do sběrnice PCI-EXPRESS).</p>
    <heading>

Řídicí jednotka 37 (řadič implementované infrastruktury obvodu FPGA).</heading>
    <heading>

Průmyslové použití</heading>
    <p id="p0032" num="0032">Dvouparametrický spektrometrický systém (předmět průmyslového vlastnictví - PPV) lze použít v jaderné energetice, průmyslové výrobě radionuklidů, radioterapii, třídění surovin podle prvko-vého složení, bezpečnostních systémech, pro potřeby ochrany životního prostředí a řadě dalších aplikací, kde je třeba identifikovat spektrum detekovaných signálů nesoucích informaci o nejrůz-nějších fyzikálních dějích.</p>
    <p id="p0033" num="0033">V dvouparametrické konfiguraci s vhodným scintilátorem (např. organickým scintilátorem stilben, resp. detektorem podobného typu) může analogově-digitálního systém sloužit jako spektrometr neutronů a záření gama s možností separace odezev detekovaných fotonů a neutronů a s jejich následným dvojparametrickým záznamem.</p>
    <p id="p0034" num="0034">Pokud není zapotřebí identifikovat typ detekované částice, lze dvouparametrický spektrometrický systém použít v jednoparametrické konfiguraci s prakticky libovolným spektrometrickým detektorem.</p>
  </description>
  <claims>
    <claim id="c-cs-0001" num="0001">
      <claim-text>

1. Upořádání analogově-digitálního systému, vyznačující se tím, že má blok (1) detektoru a zesilovačů, zahrnující detektor (11) a předzesilovač (12), připojen na blok (2) převodníků, zahrnující první a druhou dvojici AD převodníků (21 a 22), které jsou pomocí datových sběrnic připojeny na blok (3) úpravy dat zahrnující alespoň časovač (31), přičemž blok (3) úpravy dat je dále připojen na blok (4) interface, zahrnující převaděč (41), který je napojen na server (5) zahrnující RAM paměť (51) a disk (52).</claim-text>
    </claim>
    <claim id="c-cs-0002" num="0002">
      <claim-text>

2. Upořádání analogově-digitálního systému, podle nároku 1, vyznačující se tím, že blok (3) úpravy dat zahrnuje časovač (31) připojený k bloku (32) zpracování dat, který je <!-- EPO <DP n="5"> -->
připojen k bloku (33) FIFO filer, jenž je připojen k bloku (34) generátoru paketů, jenž je přes výstup (TX) připojen na vstup (RX) bloku (35) přepínače, jenž je přes výstup (TX) připojen na vstup (RX) bloku (36) interface, jehož výstup (TX) je propojen na vstup (RX) bloku (37) řídicí jednotky/řadiče, který má výstup (TX) připojen na vstup (RX) bloku (35) přepínače, zatímco blok (34) generování paketů je připojen přes blok (35) přepínače k bloku (37) řídicí jednot-ka/řadič pro řízení přenosu dat, přičemž blok (37) řídicí jednotka/řadič je dále připojen k blokům (21) a (22) AD převodníky pro jejich řízení a k bloku (36) interface.<br/>Seznam vztahových značek:<claim-text>

1 blok detektoru a zesilovačů / Detector and amplifíer</claim-text>
        <claim-text>

11 spektrometrický senzor</claim-text>
        <claim-text>

12 předzesilovač zesilovač</claim-text>
        <claim-text>

2 blok analogově digitálních převodníků (ADC) / Analog to Digital Converters 21 první dvojice AD převodníků</claim-text>
        <claim-text>

22 druhá dvojice AD převodníků</claim-text>
        <claim-text>

3 blok úpravy dat/FPGA</claim-text>
        <claim-text>

31 časovač</claim-text>
        <claim-text>

32 blok zpracování dat</claim-text>
        <claim-text>

33 FIFO fíller</claim-text>
        <claim-text>

34 blok generátor paketů /paketizer</claim-text>
        <claim-text>

35 přepínač</claim-text>
        <claim-text>

36 interface</claim-text>
        <claim-text>

3 7 řídicí j ednotka/řadič</claim-text>
        <claim-text>

4 blok interface /Combo card</claim-text>
        <claim-text>

41 převaděč</claim-text>
        <claim-text>

5 blok serveru</claim-text>
        <claim-text>

51 RAM (operační paměť)</claim-text>
        <claim-text>

52 disk;<br/>MDIO (Management Data Input/Output) - sériové rozhraní definované standardem Ethernet IEEE 802.3<br/>TX (Transmitter) - označení výstupního bodu vysílače<br/>RX (Receiver) - označení vstupního bodu přijímače<br/>LVDS (Low-voltage Differential Signaling) - standard definující elektrické parametry sériového<br/>diferenčního komunikačního protokolu<br/>XAUI (Extended Attachment Unit Interface) - standard pro rozšíření interfejsu XGMII (10<br/>Gigabit Media Independent Interface) mezi vrstvami MAC a PHY standardu komunikačního protokolu 10 Gigabit Ethernet (lOGbE) SATA (Seriál Advanced Technology Attachment) - standardní sběrnice číslicových systémů pro<br/>připojení velkokapacitních paměťových zařízení jako jsou pevné nebo optické disků SPI (Seriál Peripheral Interface) - interní sériové rozhraní komponent číslicových systémů PCI express (Peripheral Component Interconnect Express) - standard sériové systémové sběrnice<br/>číslicových systémů (náhrada starších standardů y PCI, PCI-X a AGP).</claim-text>
      </claim-text>
    </claim>
  </claims>
  <drawings>
    <figure num="">
      <img he="83" wi="164" file="0001.jpg" img-format="jpg" img-content="drawing" inline="no"/>
    </figure>
    <!-- EPO <DP n="7"> -->
    <figure num="">
      <img he="97" wi="164" file="0002.jpg" img-format="jpg" img-content="drawing" inline="no"/>
    </figure>
  </drawings>
</ep-patent-document>
